Spartan FRX Project Reproducibility Study
|ClinicalTrials.gov Identifier: NCT01676298|
Recruitment Status : Completed
First Posted : August 30, 2012
Results First Posted : January 7, 2013
Last Update Posted : January 7, 2013
|Condition or disease||Intervention/treatment|
|Analytical Reproducibility of a Medical Device||Device: Spartan FRX CYP2C19 Test System|
The FRX system is comprised of hardware and consumable components. The hardware components of the system include an Analyzer (thermal cycler with fluorescence detection capability), a notebook computer and a printer. The consumable component of the FRX system is a sample collection kit. Each kit contains a buccal swab (used to collect the patient sample) and a tube containing the reagents required for genomic DNA extraction and PCR (polymerase chain reaction) amplification stages of the test.
The Spartan FRX System is capable of detecting three CYP2C19 SNPs(single nucleotide polymorphism) (*2, *3, *17) in each test performed. An individual sample collection kit is required for each SNP tested; therefore three sample collection kits are required for each test performed on the system.
To perform a test, the user collects three buccal samples from the patient and then inserts a sample into each of the three reagent tubes (one for each of the CYP2C19 loci *2, *3 and *17). The reagent tubes are placed into the Analyzer and the FRX system automates the processes of DNA extraction, PCR amplification, fluorescent signal detection and data analysis. The system provides the user with a printed result listing the patient genotypes at the *2, *3 and *17 loci.
The objective of the study is to evaluate the performance of the FRX System under multivariate conditions. Specifically, the following variables will be included in the study:
- Test site - x3
- Operator - x6 (2 per site)
- Day - x15 (5 non-consecutive days per site)
- FRX System - x16
Test performance is defined as the number of correct genotype calls, expressed as a percentage of the total number of tests performed on the system.
For both the first-pass and second-pass results, 1-sided 95% confidence lower limits will be calculated using the score method for the % correct calls (i.e. % agreement).
Genotype results from the FRX system will be compared with results of DNA sequencing. The result of the FRX System test will be determined to be correct if the genotype calls for all three SNPs are identical to the genotypes determined by DNA sequencing for that sample/individual.
Results of the Reproducibility Study will be acceptable if the lower bound of a 1-sided 95% confidence limit of the total correct call rate per genotype is greater than or equal to 95%, based on second-pass results.
|Study Type :||Observational|
|Actual Enrollment :||8 participants|
|Official Title:||Study of the Analytical Reproducibility of the Spartan FRX CYP2C19 *2,*3 and *17 Genotyping System.|
|Study Start Date :||August 2012|
|Actual Primary Completion Date :||September 2012|
|Actual Study Completion Date :||November 2012|
- Percentage of Correct Calls to Assess Reproducibility of the Spartan FRX CYP2C19 System. [ Time Frame: After second pass result is complete (~3h) ]
Reproducibility was calculated as a percentage of the correct calls over the total calls made for each genotype group. All calls were made using the Spartan FRX CYP2C19 genotyping diagnostic system.
All data analyses was qualitative, based on the genotype calls determined by the FRX system (using on-board automated data analysis). A printed result listing the genotype call for each SNP was generated by the FRX system at the end of each run. If the result of a test is "Inconclusive" for one or more SNPs, the test were immediately repeated for the corresponding SNP(s) only, per the instructions for use. Results are reported based on both first-pass and second-pass (i.e. repeated test). For both the first-pass and second-pass results, 1-sided 95% confidence lower limits were calculated using the score method for the % correct calls (i.e. % agreement).
Please refer to this study by its ClinicalTrials.gov identifier (NCT number): NCT01676298
|Children's Hospital of Eastern Ontario|
|Ottawa, Ontario, Canada, K1H 8L6|
|Ottawa Hospital Research Institute|
|Ottawa, Ontario, Canada, K1H 8L6|
|Mount Sinai Services|
|Toronto, Ontario, Canada, M5G 1Z5|
|Principal Investigator:||Chris JE Harder, PhD||Spartan Bioscience|
|Principal Investigator:||Azar Azad, PhD||Mount Sinai Hospital, New York|
|Principal Investigator:||Marc Desjardins, PhD||Ottawa Hosptial Research Institute|
|Principal Investigator:||Jean McGowan-Jordan, PhD||Children's Hospital of Eastern Ontario|