Performance Evaluation of Blood Glucose Monitoring Systems (Polaris)
|Study Design:||Allocation: Non-Randomized
Endpoint Classification: Efficacy Study
Intervention Model: Single Group Assignment
Masking: Open Label
Primary Purpose: Diagnostic
|Official Title:||Performance Evaluation of BGStar and iBGStar in Terms of Accuracy (Polaris), Intra-assay Precision, and Hematocrit Interference (Helios Substudy) Following ISO15197 and TNO Guidelines|
- Accuracy of the blood glucose meter in comparison to a standard reference [ Time Frame: within 10 min up to 300 min after start of experiment ] [ Designated as safety issue: No ]After start of the experiment, blood glucose will be measured by means of the test devices and blood will be drawn for assessment of venous blood glucose concentration with a standard laboratory reference method
- Intra-Assay precision [ Time Frame: within 10 min up to 300 min after start of experiment ] [ Designated as safety issue: No ]Blood samples from 5 individuals with different levels of blood glucose were each measured with 10 meters of both investigational devices, the BGStar® and iBGStar®, ten times resulting in a total number of 500 measurements.
- Hematocrit Interference (Helios) [ Time Frame: within 10 min up to 300 min after start of the experiment ] [ Designated as safety issue: No ]One patient provided blood and artificial samples are generated from venous heparinized blood. These blood samples were adjusted to five different BG levels (2.8 mmol/L, 5.6 mmol/L, 8.3 mmol/L, 13.9 mmol/L, 19.4 mmol/L) at five different Hct levels (35%, 40%, 45%, 50%, 55%) to fulfill the range required by the TNO 2001 and tested by all devices.
|Study Start Date:||January 2011|
|Study Completion Date:||June 2011|
|Primary Completion Date:||April 2011 (Final data collection date for primary outcome measure)|
Diagnostic assessment of blood glucose by means of different devices
Other: Sugar infusion
Infusion of glucose to achieve high blood glucose levels
Other Name: 5 % glucose solution
Experimental: insulin infusion
infusion of insulin to achieve low glucose levels
Drug: insulin infusion
i.v. infusion of insulin
Other Name: insuman rapid
The main study (Polaris) investigated blood glucose measuring accuracy and intra-assay precision while the substudy (Helios) investigated potential haematocrit interference of sanofi-aventis BGMSs BGStar® and iBGStar® and partly of competitor BGMSs according to ISO 15197  and TNO 2001 Quality Guideline , respectively.
The primary objective of the system accuracy evaluation is to identify the number of blood glucose readings of both sanofi-aventis devices within the ISO 15197 area of acceptable accuracy of ± 20% [DEV%] for blood samples with glucose concentration > 75 mg/dL (4.2 mmol/L) and within the ISO 15197 area of acceptable accuracy of ± 15 mg/dL (± 0.83 mmol/L) for blood samples with glucose concentration < 75 mg/dL (4.2 mmol/L) as compared to the StatStrip® Connectivity reference method. The primary objective of the haematocrit interference evaluation is to demonstrate that the BGStar® and iBGStar® devices meet the Hct interference requirements according to TNO 2001 Quality Guideline when used to measure BG of venous whole blood as compared to a venous plasma reference method (YSI 2300 STAT Plus™) at Hct levels between 35 - 55% in terms of:(1) a mean deviation not exceeding ± 10% [DEV%] for hyperglycaemic glucose concentrations (150 - 180, 250 - 280, 320 - 350 mg/dL) and (2)a mean deviation not exceeding ± 18 mg/dL (1.0 mmol/L) [DEVtotal] for hypo- and normoglycaemic glucose values (50 - 60, 100 - 120 mg/dL) compared to the reference method at Hct levels between 35 - 55%.
Please refer to this study by its ClinicalTrials.gov identifier: NCT01504620
|IKFE - Institute for Clinical Research and Development|
|Mainz, Germany, 55116|
|Principal Investigator:||Andreas Pfützner, MD, PhD||IKFE Institute for Clinical Research and Development|