An Inpatient Performance Evaluation of a New Subcutaneous Glucose Sensor
The purpose of the study is to assess the performance of a new subcutaneous glucose sensor over a seven-day sensor life when used with currently marketed Medtronic Diabetes devices. In addition performance will be calculated for use of the new sensor with proposed new devices using new calibration algorithms.
Type 1 Diabetes Mellitus
Type 2 Diabetes Mellitus
Device: Sensor wear
|Study Design:||Endpoint Classification: Safety/Efficacy Study
Intervention Model: Single Group Assignment
Masking: Open Label
Primary Purpose: Basic Science
|Official Title:||An Inpatient Performance Evaluation of a New Subcutaneous Glucose Sensor|
- Glucose Sensor Accuracy When Compared to Laboratory Standard (YSI): Proportion of Glucose Sensor Readings That Met Accuracy Criteria [ Time Frame: Days one through six of sensor use ] [ Designated as safety issue: No ]The primary accuracy parameter (primary effectiveness endpoint) was the comparative readings of paired sensor and YSI glucose readings, measured on days 1 through 6. Accuracy is defined as within 20% agreement between YSI and paired sensor (within 20 mg/dL if YSI <80 mg/dL). Accuracy ranges from 0 - 100, with higher number suggests better accuracy.
- Device Related Moderate or Device Related Severe Adverse Events [ Time Frame: days one through six of sensor wear ] [ Designated as safety issue: Yes ]
Device related moderate adverse event: low level of inconvenience or concern to the subject and may interfere with daily activities but is usually improved by simple therapeutic remedy.
Device related severe adverse event: interrupts a subject's daily activity and typically requires intervening treatment.
Note: device related determination is made by the site that there is a reasonable possibility that the adverse event may have been caused by the device.
|Study Start Date:||April 2010|
|Study Completion Date:||November 2010|
|Primary Completion Date:||November 2010 (Final data collection date for primary outcome measure)|
Active Comparator: Sensor
All subjects that wear sensors (all subjects)
Device: Sensor wear
All subjects to wear sensors
Other Name: Enlite Sensor
The first generation Medtronic MiniMed Subcutaneous Glucose Sensor (Sof-Sensor) was originally approved by the FDA for commercialization as part of the Continuous Glucose Monitoring System (CGMS) on June 15, 1999 (PMA 980022). The Sensor is composed of a microelectrode with a thin coating of glucose oxidase beneath several layers of biocompatible membrane. This same sensor was used as part of subsequent continuous glucose monitoring (CGM) systems, such as the Guardian REAL-Time and Paradigm REAL-Time sensor augmented insulin pump. The current configuration of sensor has undergone in vitro and in vivo testing. A new second-generation glucose sensor (herein referred to as the Comfort Sensor) has been developed. The first-generation glucose sensor was approved with reported sensor accuracy (MAD) of 18%; it was labeled for a maximum use duration of 72 hours, using only the abdomen as an insertion site. The new sensor is shorter and has a smaller diameter, with a smaller gauge introducer needle. The new sensor inserter is designed to be used with the new sensor. The objectives of this study are to 1) Assess performance of the Comfort Sensor when used over a period of seven days with currently available devices, and 2) Assess performance of the Comfort Sensor when used over a period of seven days using new calibration algorithms (post-processed with algorithm for future devices). Accuracy data will be calculated based on comparing calibrated glucose sensor values to a "gold standard" (YSI plasma glucose values) in adult subjects during in-clinic testing, and on comparing calibrated glucose sensor values to glucose meter values during in-home testing. The devices that will be used for gathering sensor data during this study will be: 1) the Comfort Sensor, 2) the MiniLink (Transmitter), 3) the Guardian REAL-Time Display Device and 4) the CGMS iPro.
Please refer to this study by its ClinicalTrials.gov identifier: NCT01112696
|United States, California|
|AMCR Institute, Inc.|
|Escondido, California, United States, 92026|
|Sansum Diabetes Research Institute|
|Santa Barbara, California, United States, 93105|
|United States, Washington|
|Rainier Clinical Research Center, Inc.|
|Renton, Washington, United States, 98057|
|Study Director:||Scott Lee, MD||Medtronic Diabetes|