Comparison of the Video and Macintosh Laryngoscope in Patients Who May be Difficult to Intubate
|Intubation, Endotracheal||Device: DCI Video Intubation System-Macintosh Laryngoscope Device: Macintosh Laryngoscope||Phase 4|
|Study Design:||Allocation: Randomized
Intervention Model: Parallel Assignment
|Official Title:||A Comparison of Laryngoscopy Techniques Using the Video Laryngoscope and the Traditional Macintosh Laryngoscope in Patients Who May be Difficult to Intubate|
- 5 scale score of glottic view; Time and number of attempts required; Level of difficulty; Degree of irritation of the pharynx, epiglottis, and aryteniods
- Vital signs, oxygen saturation, and end-tidal carbon dioxide
|Study Start Date:||August 2003|
|Study Completion Date:||February 2007|
|Primary Completion Date:||February 2007 (Final data collection date for primary outcome measure)|
Despite improvements made to the traditional laryngoscope blade since its invention, occasionally intubation of the trachea cannot be accomplished with facility, even in patients with anatomy that does not predict difficult intubation. It is estimated that endotracheal intubation is performed on some 8 million patients per year in the United States. Of these endotracheal intubations, approximately 80% are performed by direct laryngoscopy with transoral placement of the endotracheal tube (ET) into the trachea. There is fairly uniform reporting of the incidence of failed intubation in the literature; it occurs in approximately 0.05% or 1:2230 of surgical patients and in approximately 0.13% to 0.35%, or 1:750 to 1:280, of the obstetric patients. The incidence of unsuspected difficult intubation is estimated to be higher at 3%. One factor that contributes to difficult intubation is poor visualization.
The VL is designed to optimize visualization by presenting to the operator an enlarged video image of airway structures. In contrast, using conventional laryngoscopy, anesthesiologists have only a "keyhole" view of the airway structures; a view that may be further obscured during attempts to pass the ET.
The VL consists of a laryngoscope handle and Macintosh blade that have been modified to provide a video image of airway structures on a screen, which can be conveniently located directly in front of the anesthesiologist. A micro video module is contained in the modified handle. An image/light bundle is introduced into the standard blade. This system has been previously tested in which the consensus was that the device is extremely easy to learn to use because most anesthesiologists are familiar with the use of the Macintosh blade. It has also been useful in the instruction of laryngoscopy by non-anesthesiologists.
There are several potential advantages of a video image in the context of direct laryngoscopy. The system provides high quality video images that are enlarged on the video monitor for easier visualization. If laryngeal manipulation is required to improve visualization of laryngeal structures, the intubator and the person assisting can coordinate movements as they observe, simultaneously, the image on the video monitor. With the video image projected from the distal end of the laryngoscope blade, laryngeal structures are kept in view as the ET is passed through the oropharynx into the trachea.
Comparison: Video Laryngoscope as a conduit for possible difficult intubation compared to the traditional Macintosh blade.
Please refer to this study by its ClinicalTrials.gov identifier: NCT00178555
|United States, Texas|
|Memorial Hermann Hospital|
|Houston, Texas, United States, 77030|
|Principal Investigator:||Carin A. Hagberg, M.D.||The University of Texas Medical School at Houston|